Custom Search
Over 10000 Pages Indexed

Your Host 
Kirt Blattenberger ... single handedly redefining what an engineering website should be.
5CCG (5th MOB): 
Hobby & Fun
Airplanes and Rockets: Equine
Kingdom: 
•−• ••−• −•−• •− ••−• •
RF Cafe Morse Code >Hear
It<


Job Board 

About RF Cafe© 

RF Cafe EMail 
May 1934 Radio News and ShortWave[Table of Contents]These articles are scanned and OCRed from old editions of the Radio & Television News magazine. Here is a list of the Radio & Television News articles I have already posted. All copyrights (if any) are hereby acknowledged. 
Any problem involving proportion can
be solved graphically without much difficulty. Such problems
occur often in a serviceman's and experimenter's life; therefore,
the chart, presented this month should prove applicable in many
cases. For instance, one can use any milliammeter as an ohmmeter but then it takes a little arithmetic to find the resistance after readings are taken. This arithmetic will be eliminated when using the chart in Figure 5. The chart can also be applied to the slidewire bridge, a construction is given for an ohmmeter scale employing a milliammeter in the two popular circuits. In the February 1932 issue of Radio News, the formulas for the different kinds of charts were derived and here the Nchart was introduced. It consists of three linear scales in the form of an N or a Z and is useful in several kinds of proportionproblems and some forms of multiplication. As seen in Figure 1, the chart has the drawback that it is hard to utilize the extreme end of the diagonal scale since the line, 1, does not intersect scale c within the limits of the paper. In some cases, it will be found that nearly one fifth of the scale b on each end cannot be used. Therefore. instead of using the N chart, a logarithmic arrangement was used in Figure 5, even though this meant the calculation of a special scale. It solves the same formula as the N chart. Now let us proceed with the problem. The ordinary ohmmeter is usually something like the circuit in Figure 2. The 01ma. meter is the most popular, and therefore it is used here in the examples, but any milliammeter can be used. In fact. for lowresistance measurements, an 050 m. meter is very useful. R is usually made so that the meter shows fullscale reading when the test prods are shortcircuited. This, too, is not essential, but it is desirable since it gives a longer scale and it simplifies computation. Suppose that we have such an ohmmeter, in which the meter shows fullscale deflection when the prods are shortcircuited, and that it shows a lower reading, m, when the resistance x is inserted. Then, Here the numeral 1, represents the fullscale reading (1 ma.) If a different kind of meter is used with a full scale reading of n ma., then, Referring to the chart of Figure 5, connect the meter reading on scale M (A) with the value of R on scale R and the intersection on scale X shows the value of the unknown resistance. For instance. if, as in our case, R is 4500 ohms and the meter went down to .2 ma. with the resistance X inserted, then the sample line shows that the resistance of X is 18000 ohms. All you have to know is the value of R. This is the resistance which will let just enough current flow to make the meter show fullscale reading. It can be figured out with Ohm's law. With a 1ma. meter and a 4.5volt battery, R is 4500 ohms. With a 10ma. meter and a 4.5volt battery it would be 450 ohms and with a 1ma meter and a 1.5volt battery it would be 1500 ohms, etc. The range of such an ohmmeter is from 1/10 R to 10 R. It is possible to obtain some kind of reading above and below this, but they are not accurate. 
The second type of ohmmeter has the advantage
of being especially suited to the measurement of low resistances
without drawing a very heavy current. This is illustrated in
Figure 3. The unknown resistance is shunted across the meter,
which also makes the meter go down from fullscale deflection.
For higher resistances it may be placed across the meter, plus
a resistance, a, in series with it. The sum of a, and the resistance
of the meter we shall call R. Then the value of X is where m is the reading of the meter with X connected to the test prods and n is the reading when the test prods do not touch each other. The same chart (Figure 5) can be used again, but for this type of ohmmeter use scale M (B). The sample line shows that if the meter went down to .8 mils and the resistance of the meter was 45 ohms, the resistance of X would be 180 ohms. You can extend the ranges of R and X by adding or subtracting the same number of zeros to both A scale can be constructed for a given ohmmeter rather easily. This is illustrated in Figure 4. Line L represents the scale on the meter. The same divisions and numbers should be put on it. Then, in order to get the ohmmeter divisions, draw two lines, a and b, through the ends of the scale at any convenient angle as shown, a and b are parallel. On a, regular divisions are set off representing ohms, on any convenient scale. Next you should find a point P on scale B so that OP equals R when measured in the unit employed on a. Then draw lines from P to the divisions on a. The intersections give the divisions for the ohmmeter scale on L. This construction is good for the series ohmmeter (Figure 2) only. For the shunt type, the divisions of a should be put on b instead and the point P should be taken on a in the same manner as before. It will be found difficult to get to both ends of the scale with just one position of P. After some divisions have been made with P in one position, the point can be shifted, which means that you are using another measuring unit on both line a and b. It will be found that the scales obtained by this method, for the ohmmeter of Figure 2, resemble those for Figure 3 if R is the same; only, they are mirror views of each other. When another range is required, it can be done by making R twice as large (or 10 times as large) and then multiplying the readings of your previous scale. In Figure 6, two examples of completed scales are given. One is for a 01ma. meter in the series circuit with a resistance of R equals 4500 ohms. Its range can be extended by employing a resistance of 45000 ohms and a 45volt battery. Then the scale readings should be multiplied by 10. The other scale is designed for an ohmmeter of the shunt type, employing an 01ma. meter with an internal resistance of 30 ohms. If your meter is somewhat less than 30 ohms you should connect some resistance in series with it so as to make the total 30 ohms. Shunting the meter with a 3.3ohm resistor will make the scale lower, multiply all values with.1. When your meter has a resistance of 50 ohms (such as the universal meter) all scale values should be multiplied by 5/3 or 1.67. 
Electronics Nomographs and Circuits
Figure 5  A Millimeter as an Ohmmeter
A Disruptive Web Presence™
Custom Search
Over 10,000 pages indexed! (none duped or pirated)

Read About RF Cafe Webmaster: Kirt Blattenberger KB3UON 
RF Cafe Software
RF Cascade Workbook 
Product & Service Directory Personally Selected Manufacturers 

RF Cafe TShirts & Mugs  
RF Cafe SoftwareCalculator WorkbookRF Workbench Smith Chart™ for Visio Smith Chart™ for Excel 