Electronic Component Checking (L, C, and R)
February 1957 Radio & Television News Article
this article on component (resistor, capacitor, and inductor) measurement was written, readily
available, inexpensive multimeters were not in existence. For about $20 you can now buy a brand new
handheld DMM that will make very accurate resistance measurements and reasonably good capacitance
measurements at frequencies up to a few MHz, where lead inductance starts to be significant (test
frequency is usually only a few kHz). Finding an affordable, accurate inductance meter is another story.
LCR meters can be purchased on eBay, but don't be surprised if the quality is not very good. The
most accurate measurement method uses a frequency in the realm of actual operation, and this article
presents methods that will allow you do do just that by using typical bench top instruments.
of Contents]These articles are scanned and OCRed from old editions of the Radio & Television News magazine.
Here is a list of the Radio & Television News articles
I have already posted. As time permits, I will be glad to scan articles for you. All copyrights (if any) are hereby
See all available
vintage Radio News articles.
N. H. Crowhurst
discussion of various ways that circuit components in radio and audio equipment can be checked without trouble.
Sometimes the simpler things one encounters in radio and audio work are apt to get overlooked. For example,
it would seem to be quite an easy matter to check the inductance of a smoothing choke or the capacitance of an electrolytic
capacitor, with the correct polarizing current or voltage. However, when one looks around to find a test instrument
to make the measurement, it just isn't readily available, so we are virtually forced into the routine of taking
things for granted.
If we wish to check as to whether a certain component is functioning correctly or not,
the only available method seems to be by substitution, using another component of the same type. Often this proves
to be somewhat unsatisfactory, because the results can be inconclusive. We really need to know how to check the
various fundamental components used in radio: resistance, inductance, and capacitance, to varying degrees of accuracy,
according to their purpose.
The simplest method of resistance
checking is by means of a simple ohmmeter, either an instrument built specifically for this purpose or an ohmmeter
range on a volt-ohm-milliammeter. Accuracy of this method of measuring resistance rarely exceeds 10% and may not
even be as good as this.
Assuming that the accuracy of the moving coil meter used for the instrument is
±2% and that the resistors used in the instrument are accurate to ±1 %, the accuracy of the instrument as a perfect
comparator between the internal and external resistances cannot be better than ±1%. And the accuracy of comparison
is only to within ±2% of the full-scale current reading on the scale. If the scale reading, on a voltage or current
scale, is compared with the reading on the ohms scale, it will be found that an error representing 2% of full scale
in voltage or current reading may amount to an 8% error in resistance value. This is at the point of maximum accuracy
of comparison, between the external resistance being measured and the internal resistance of the instrument.
Thus it is seen that the best accuracy obtainable using an instrument with a ±2% movement and ±1 % internal
resistance gives a guaranteed accuracy at center scale reading of 9%. At readings between one-third and 3 times
the resistance value, which is the range one might expect to use before switching to the next scale, the accuracy
can reasonably be expected to stay within 10%. With an instrument using lower accuracy components than those used
for illustration, the accuracy of the final reading in ohms will be considerably poorer than 10%.
it will be evident that an ohmmeter can only be used to make a rough check as to whether a resistance is within
the preferred value range for which it is color coded - if it is of a ±10% or higher tolerance rating. To check
that the resistance is within ±10% of its rated value, the result is a little doubtful and it is certainly impossible
to rely on an ohmmeter reading to check to a tolerance of ±5% or closer.
Although the ohmmeter readings
cannot be trusted for checking to close tolerances, it is possible to use an ohmmeter to check for reasonably good
matching between pairs of resistors, if this happens to be the requirement rather than close precision in actual
As an example, in many push-pull amplifiers the resistors responsible for controlling the gain in
the two halves of the push-pull arrangement must be closely matched to ensure balance. Production values may be
specified to 5% or even closer tolerances, to avoid the necessity of having to select matched pairs, but the essential
feature is that the value of the two corresponding resistors shall be within a close tolerance of one another. It
will not necessarily matter if both of them are, say, 10% or 15% from their nominal rating, as long as they are
within 5% of each other. This the ohmmeter is reasonably capable of checking, because it is quite possible to read
an ohmmeter scale to within 5%. Since the question as to whether the reading is within 5% of its actual value is
unimportant in this particular application, the significance of the reading does not matter as much as whether the
two resistors which should be matched give readings within 5% of one another.
For some applications, however,
such as calibrated attenuators or instruments for use in radio it is necessary to check resistor values to closer
limits such as 5%, 2%, or even 1%, as the case may require. In these circumstances it is important that the value
shall really be within the specified percentage of its rated value. The only method of making a measurement that
is satisfactory for this purpose is to use a Wheatstone type bridge, using calibrated elements whose accuracy is
better than the required component accuracy.
For most radio purposes the Leeds & Northrup bridge used
for telephone line work is quite accurate enough. In using a bridge there are two things that control the accuracy
of the reading obtained: (1) the accuracy of the resistance elements of the bridge itself, and careful attention
to see, that contact resistance does not contribute an appreciable fraction under any circumstances; and (2) the
sensitivity of the null detector.
This second cause of inaccurate results can be checked by unbalancing
the bridge by a known percentage to see that an adequate off-balance reading is obtained. Suppose, for example,
the value required is 120,000 ohms, ±5%. Having balanced the bridge and obtained a null at, say, 120,000 ohms, the
resistance in the calibrated arm should be altered by 5%, which represents a change of 6000 ohms.
in 6000 ohms additional in the calibrated arm shows appreciable deflection, then the reading may be regarded as
accurate; but if the addition of 6000 ohms does not produce noticeable deflection from balance on the null detector,
the result is not reliable. To improve its reliability one can either use a larger battery voltage or source of
supply to the bridge, or else get a more sensitive null indicator.
Before leaving this discussion of resistance
values it should perhaps be emphasized that it is not wise to put absolute trust in the color coding on a resistor.
Occasionally even the best resistor will be found incorrectly color coded. If the error happens to be in the third
color of the code, then the discrepancy in resistance value will be a matter of shifting the decimal point which
can be quite serious. Also with some sets of coding colors the difference between some of the colors is somewhat
difficult to determine, especially after the component has aged. For example, orange and brown can get to look quite
Usually the first and second colors in the code can be identified by the combination used, from the
recognized preferred value range. If the first color is blue, representing 6, the second color will most likely
be either red, representing 2, or gray, representing 8, because 62 and 68 are the preferred values in the 60 to
70 range. But there is no such ready clue as to the likely color of the third band: it could just as easily be brown
or orange. Thus a resistor in which this color looks at all doubtful could be either 620 ohms or 62,000 ohms, which
is a considerable difference!
This is where an ohmmeter check can easily determine which of the two values
Turning now to various kinds of inductance: the measurement of components not intended for the passage of d.c. and
without iron cores is a fairly simple matter, with the aid of a conventional inductance bridge. Using such a bridge,
employing either the Hay or Maxwell configuration (see Fig. 1), the inductance can be measured at a frequency suitable
for the purpose, with a method quite similar to the operation of a bridge for measuring resistance.
Fig. 1. Bridge configurations for measuring inductance. (A) the "Hay" bridge. (B) the "Maxwell" bridge. Relative
advantages of each type are discussed in the article.
Fig. 2. Modification of the "Hay" bridge to enable it to measure inductance with polarizing current flowing.
Care is necessary not to exceed the dissipation rating of the various bridge elements. See text.
Fig. 3. A simple inductance checker circuit for determining inductance with the polarizing current flowing in
principal difference is that two kinds of adjustment are usually necessary to achieve null, because of the necessity
for balancing the bridge in both amplitude and phase. This enables the bridge to give a reading of both inductance
value and "Q" or loss factor. Bridges of this type are clearly marked to indicate the correct setting of the controls
for making each kind of measurement.
There is usually no difficulty in achieving a null with the air-core
type of coil, but if the inductance employs any kind of core, the null may not be quite as definitive, because of
the distortion of the injected test signal caused by the core. Also, if the generator signal itself has any appreciable
harmonic content, a Hay bridge will never give a balance at both fundamental and harmonics at the same setting.
On the other hand, with an inductance where the only loss is due to its resistance, such as occurs in an air-core
coil, the Maxwell bridge will give a fairly satisfactory balance for both fundamental and the lower harmonic frequencies
at the same setting.
When measuring an inductor that employs any kind of core to increase the permeability,
the magnetizing current is liable to distort so the inductor itself will generate some harmonics not present in
the input from the generator. When the bridge is balanced to the fundamental generator input, there will be a residual
harmonic present at the null point, generated by the inductor itself.
This is a good reason for using earphones
if the generator frequency is in the audio range. Otherwise an oscilloscope with amplifier may be used as a null
detector. It is then possible, listening to the tone or looking at the trace, to determine when the fundamental
is balanced and the residue consists of harmonics.
But the conventional type of bridge is only suitable
for measuring inductances where there is no polarizing current. The usual variety of smoothing filter choke has
to provide a specified inductance when polarizing current is flowing and the inductance in the absence of such polarizing
current will be considerably higher than the rated inductance of the choke with polarizing current. Unfortunately
there is no simple fixed relationship between these two values.
If the choke has been designed to provide
its maximum inductance at the polarizing current for which it is designed, the air gap will be adjusted so that,
at this value of polarizing current, either reduction or increase of the air gap would result in a reduction of
inductance value. However, in the absence of polarizing current, increasing the air gap will always reduce inductance
value, while reducing the air gap will always increase inductance value.
From this simple fact it is evident
that measuring an inductance with no polarizing current flowing is no criterion of its performance with polarizing
current. It can, of course, provide a check that the inductance is not completely missing, due to short-circuited
turns, in which case the inductance might not even be adequate without polarizing current flowing. But the fact
that the inductance may measure twice its required value with polarizing current is no evidence that the choke will
give its rated value with polarizing current.
Fortunately, with filter chokes of this nature close tolerances
are not too important. Usually a compliance with a minimum inductance value will suffice.
It is sometimes
possible to use a modified Hay bridge, as shown at Fig. 2, to inject a polarizing current so as to measure the inductance
with the polarizing current flowing. But this can be a dangerous procedure, because the polarizing current may exceed
the wattage rating of some of the internal components of the bridge and cause permanent injury to it. It is, therefore,
better to devise a simple checking arrangement, as shown schematically in Fig. 3.
This does not employ a
bridge method, but checks the inductance by injecting a known frequency and comparing the a.c. voltage developed
across the inductor with that across the resistor in series with it. The relation between the a.c. components of
voltage developed will enable the approximate inductance value to be calculated. This does not take into account
the effect of the inductor distorting the waveform of the a.c. signal component, which invariably occurs in this
type of inductor and is, in fact, another reason why any attempt to produce a precise figure of inductance will
be somewhat meaningless. A rough check of this nature is quite adequate for the purpose.
If 60 cycles is
the supply frequency for the a.c. component, dividing the calculated impedance of the inductor by 377 will give
the inductance value. For example, suppose the series resistor used is 100 ohms (carefully checked in value), and
the a.c. voltages measured across the resistor and inductor are 2 and 30 volts, respectively: then the impedance
of the inductor at 60 cycles is 1500 ohms, representing approximately 4 henrys.
For measuring all except electrolytic capacitors there are two methods, which correspond in relative accuracy with
the ohmmeter and bridge methods used for measuring resistance.
Fig. 4. The "Drysdale" bridge which is used for measuring capacitance. Refer to text.
Fig. 5. A simple bridge for capacitor checking that forms the basis of a number of commercial units on the market.
The null detector is usually a "magic eye" tube.
Fig. 6. Modification of a "Drysdale" bridge to permit the measurement of electrolytic capacitors with polarizing
Fig. 7. Modification of the simple bridge of Fig. 5 to enable polarizing voltage to be applied to the electrolytic
The Drysdale bridge (see Fig. 4) is a modified
Wheatstone bridge, in which resistance arms are used in the ratio positions, while a calibrated decade capacitor
is substituted for the calibrated resistance in the variable standard arm. This type of instrument can give capacitance
results comparable to those obtained with the Wheatstone or Leeds & Northrup bridge for resistance, but its
use involves careful adjustment of a number of controls until a null is achieved.
The alternative method
of capacitance measurement also uses a bridge, but one in which the null is much more quickly achieved. In this
bridge (see Fig. 5) a standard capacitor is used in one arm, the unknown capacitor in another arm, and a single
potentiometer-type resistance for the other two arms. This resistance is calibrated on the basis of the ratio between
the unknown and standard capacitors necessary to achieve null.
With this type of bridge the unknown capacitor
is connected across the terminals of the bridge and the one dial turned until null is indicated. The capacitance
value is then read off the dial. The accuracy of this type of instrument is usually comparable to that of an ohmmeter,
depending upon the accuracy with which the potentiometer type resistance has been calibrated.
these methods is really satisfactory for the measurement of electrolytics. This can better be understood by discussing
a little further the behavior of electrolytic capacitors under different conditions.
In the first place,
electrolytic capacitors freshly formed ready for use, have a dielectric film on the active plate of the correct
thickness for the working voltage. Under this condition the capacitor should have its rated capacitance.
But if the capacitor is operated consistently at a lower polarizing voltage, the thickness of the formed film
will gradually deteriorate with the result that the effective capacitance will increase somewhat. This is not necessarily
detrimental to the performance of the capacitor, provided it is not subsequently required for service at its nominal
In much the same way electrolytic capacitors kept in storage also show a deterioration
in the dielectric film resulting in an increase in effective capacitance. This means if a six-month-old capacitor
is taken from the shelf and measured on a regular capacitance bridge, without applying the necessary polarizing,
it will probably show a value considerably in excess of its nominal value. However, it will not be satisfactory
for operation until the electrolytic film has been formed up to the requisite thickness for its working voltage.
This will have to be done with the aid of a limiting resistor connected in series with the capacitor to
limit the polarizing current while the film is forming. Only when the film has formed up so the voltage appearing
across the capacitor is at its working value without excessive leakage current can its capacitance be measured to
give a reliable indication of its operating condition.
Also, if the capacitor is to be installed in a piece
of equipment for operation at its nominal working voltage, it is vital that this reforming of the capacitor be performed
before installation, so the capacitor does not take an abnormally high leakage current when the power is switched
on and possibly destroy itself before it has had a chance to become correctly reformed.
The correct measurement
of electrolytic capacitors with polarizing voltage applied can be undertaken with either type of bridge, modified
to a certain extent, as shown in the schematics of Figs. 6 and 7. If the actual capacitance value of an electrolytic
capacitor is not vital, which often is the case, then all that is necessary in installing a new one is to ensure
that it is correctly formed to its working voltage before connecting it in. This may be done with the aid of the
circuit shown in Fig. 8, which consists of a high resistance feeding the capacitor with a voltmeter across it to
indicate when working voltage has been reached. The resistor limits the leakage current through the capacitor to
well within the maximum leakage current allowed, and when the capacitor has reached its nominal charged voltage,
it can then be removed from the charging arrangement. Then, after discharging the capacitor for the sake of safety,
the capacitor is ready for installation in its intended circuit. Discharge should preferably be accomplished through
a fairly large resistor. The common practice of short-circuiting a fully charged capacitor results in a very high
discharge current that may damage the capacitor.
Sometimes a capacitor which has been in stock a long time
will deteriorate in the quantity of electrolyte present, so the capacitance will fall low in value, even after it
has been adequately reformed.
If it is not convenient to build a capacitance measuring arrangement incorporating
the polarizing supply, a fairly legitimate result can usually be achieved by ensuring that the capacitor is correctly
formed using the polarizing jig of Fig. 8, then discharging the capacitor and finally measuring it immediately with
the aid of one of the conventional capacitance bridges without polarizing voltage.
If the electrolytic capacitor
is reasonably stable, a null will be obtained which will not vary at a perceptible rate. If the capacitor is not
sufficiently stable to be reliable in use, the null may be observed to vary perceptibly while the measurement is
being taken. If the capacitance varies at a rate that can be noticed while making the measurement, then the capacitor
should be discarded as insufficiently stable for reliable operation.
The foregoing discussion has covered
the more common measurements necessary on resistance, inductance, and capacitance. Sometimes much more precise methods
of measurement are necessary, especially where the equipment is for some kind of standard operation such as a precision
oscillator. In this kind of application it is often necessary to make measurements, not only as to the precise value
at room or ambient temperature, but to determine the effect of temperature on the component. To make such measurements,
only precision bridge apparatus is satisfactory, and the component should be measured under carefully controlled
conditions of temperature and the measurements repeated at different temperatures, to discover what temperature
coefficient the component possesses. Fig. 8. Details for constructing a simple jig for forming electrolytic capacitors
up to their working voltage. See article.
Fig. 8. Details for constructing a simple jig for forming electrolytic capacitors up to their