RF Cafe Software
About RF Cafe
1996 - 2016
BSEE - KB3UON
RF Cafe began life in 1996 as "RF Tools" in an AOL screen name web space totaling 2 MB. Its primary purpose was to provide me with ready access to commonly needed formulas and reference material while performing my work as an RF system and circuit design engineer. The Internet was still largely an unknown entity at the time and not much was available in the form of WYSIWYG ...
All trademarks, copyrights, patents, and other rights of ownership to images and text used on the RF Cafe website are hereby acknowledged.
My Hobby Website:
Try Using SEARCH
to Find What You Need.
There are 1,000s of Pages Indexed on RF Cafe !
October 1958 Popular ElectronicsTable of Contents
People old and young enjoy waxing nostalgic about and learning some of the history of early electronics. Popular Electronics was published from October 1954 through April 1985. All copyrights are hereby acknowledged. See all articles from Popular Electronics.
Commercially available test equipment for hobbyists in the 21st century is pretty high quality at a relatively low price, so the motivation for buying parts and building your own RF power meter has to be driven not by cost, but by a desire to gain the experience. This article presents a simple RF power meter that can be built for about $25 worth of parts. All the parts should be readily available except for the Raytheon CK-721 (or CK-722) germanium PNP transistor. The 2N3906 and the 2N2907 have been suggested as replacements, but some changes in the biasing resistors might be required.
See all articles from Popular Electronics.
Simple R.F. Meter
By Charles J. Schauers
Measure r.f. output of the transmitter in your ham shack ... or use this meter in -a dozen other ways
This inexpensive r.f. indicator has a wide variety of uses around the ham shack or mobile radio installation. It can act as an absorption frequency meter (if calibrated), a field strength meter, neutralization indicator, or modulation monitor with phones. However, the main job of the model shown is to indicate proper antenna loading for my "minified" mobile transmitter.
It is relatively simple to put together, and nearly any low-priced transistor will work well. However, for maximum sensitivity, a transistor with a beta (current amplification) of .between 25 and 45 should be used.
Construction. A 3 3/4" x 3" x 2 1/8" aluminum chassis houses all parts. The coil (L1) is tapped and connected. to switch S1 before installation in the box to facilitate soldering. Then the coil is cemented to the box by its plastic support.
Care should be taken in soldering the crystal diode (CR1) into the circuit by "heat sinking" the connections with a pair of long-nose pliers. A socket should be used for the transistor.
Fasten the 1 1/2-volt cell to the chassis with household cement. With normal use, it should last almost its shelf-life.
After the unit is turned on, zero the meter with potentiometer R3 in the collector circuit. Attach a small wire to the input binding post on the rear of the box which feeds r.f. to the tuned circuit, and you are in business.
Applications. If the device is to be used as an absorption frequency meter, it can be calibrated with a heterodyne frequency meter coupled to the input post through a small (about 500-μμfd.) capacitor.
A 2-1/2" length of wire is sufficient for r.f. pickup when checking oscillator, doubler or buffer and final amplifier stages of a transmitter.
When using the indicator as a field strength meter to adjust a beam antenna, the pickup wire length will depend upon the distance from the antenna and how much power is being applied from the transmitter's final amplifier. Usually, a 2" piece of wire will afford sufficient pickup at 100 feet with the average low-power transmitter when the device is hand-held.
To provide some attenuation of very strong signals, the indicator can be used harmonically. Set the bandswitch to 40 meters when you want to measure carrier strength on 80 meters.
As a means for tuning mobile or fixed transmitters (especially those employing pi-output-networks), this unit enables one to determine very quickly if the antenna and not the pi-network is taking the load. For mobile operation, the regular auto broadcast antenna can be used for signal pickup. However, the device should be harmonically operated as described above because of the strong signal present. If the auto antenna is not used, try a small piece of insulated wire taped to the inside of the front bumper, connected with shielded wire to the indicator.
If you are interested in monitoring your modulation, a pair of magnetic phones can be connected in the collector circuit of the transistor. The meter is disconnected (as well as the top of the balancing potentiometer) and the phones are connected between battery minus and collector.
B1 - 1.5·volt D cell
C1 - 140·μμfd. miniature variable capacitor
C2 - 0.002·μμfd. mica capacitor
CR1 - Crystal diode (Sylvania 1N34 or equivalent)
L1 - #24·wire, 1"·diameter coil (Airdux 832T or B & W 3016 - 32 turns per inch, tapped as shown in schematic)
M1 - 0-1 ma. d.c. meter
R1, R2 - 680·ohm, 1/2·watt resistor
R3 - 6500-10,000 ohm slotted shaft potentiometer
S1 - 1-p., 4-pos. rotary switch
S2 - S.p.s.t. toggle switch
TR1 - Transistor (General Transistor GT·87 or GT-88
or Raytheon CK·721)
1-Aluminum chassis box (LMB·135)
HOW IT WORKS
Radio frequency energy tuned by L1-C1 is applied to diode CR1. The rectified current then takes a path through the base-emitter circuit of transistor TR1.
Current amplification occurs and is read by the 0.1 milliammeter. Capacitor C2, connected between the transistor base and ground. bypasses the radio frequency. The greater the strength of the r.f. signal picked up, the higher the reading on the meter.
Posted September 19, 2011