[Table of Contents]People old and young enjoy waxing nostalgic about
and learning some of the history of early electronics. Popular Electronics was published from October 1954 through April
1985. As time permits, I will be glad to scan articles for you. All copyrights (if any) are hereby acknowledged.
RF Cafe visitor Terry W. recently wrote to me about a topic that ended up mentioning RCA's vacuum
tube voltmeter (VTVM) known as the VoltOhmyst. VTVMs were the era's high-input-impedance
multimeter, before field effect transistors (FETs) came on the scene. The higher a meter's
input impedance, the less it loads the circuit under test. Any meter connected across a
circuit appears as a parallel load to the source, so the closer it is to an open circuit, the
better. Terry mentioned how the VoltOhmyst was a key component on test benches of many
professionals. You can still buy various models of the
See all articles from
Popular Electronics Builds a Vacuum-Tube Voltmeter
board makes the RCA VoltOhmyst an easy-to-assemble kit
The first piece of
quality electric test equipment you should have for your test bench is a vacuum-tube voltmeter.
A VTVM found in service shops and industrial plants throughout the country, the RCA "VoltOhmyst,"
Type WV-77EK, is now being offered in kit form.* What It Can Do.
VoltOhmyst WV-77EK measures a.c. (r.m.s.) and d.c. voltages up to 1500 volts, peak-to-peak voltages
to 4000 volts, and resistance up to 1000 megohms. There is a high input impedance on all d.c.
and a.c. voltage ranges, allowing the use of this VTVM in circuits where VOM's with a lower input
impedance would result in loading of the circuit under test - and a resultant error in voltage
The VoltOhmyst utilizes a push-pull balanced d.c. bridge with the meter in the
plate circuit, which affords excellent linearity of response, good stability, and very high input
impedance. Additional features include: provision for zero-center indication, useful in discriminator
and bias measurements; separate scales for low a.c. voltage measurements to assure accurate readings;
a circuit design which allows measurement of a.c. in the presence of d.c. and vice versa; a separate
d.c. probe with a 1-megohm resistor which minimizes capacitance-loading effects; and electronic
protection against meter burn-out. Also, the resistors in the ohmmeter ranges are protected by
a separate fuse.
* For complete information on the VoltOhmyst, write to Radio Corporation
of America, Commercial Engineering Dept., Section PE-10, Harrison, N. J.
Putting It Together.
Large, clear meter face
permits quick, accurate readings.
unit. Note the three internal adjustment potentiometers. Diagram
below is from the construction sheet.
Well planned illustrations
reduce complex wiring to a simple task.
The WV-77EK utilizes a printed-circuit board to facilitate
assembly. This board provides a rugged, pre-wired mounting for the components and, if the parts
are properly inserted and soldered, makes for a neat and trouble-free assembly.
number of the part to be mounted is printed on one side of the board, and the copper wiring is
etched on the other side. When the assembly instructions call for mounting a part, make sure the
leads of the component form a right angle to the body of the part. The leads, when properly bent,
form the two long portions of a "U" shaped unit, the bottom of the "U" matching the dimension
between the two holes. Some parts, such as the disc capacitors, tube sockets and the selenium
rectifier, do not require bending.
Place the leads in the holes provided and pull the
part snug to the board, so that the leads protrude on the etched side of the board. Spread the
leads slightly to prevent the part from falling out. After soldering the leads to the copper foil,
cut them to 1/8" from the board. Recommendations.
The instructions for
assembling and wiring the VoltOhmyst have been carefully thought out and presented in seven major
construction steps. The POPULAR ELECTRONICS editor who assembled the VoltOhmyst makes the following
- Insert R14 as the first substep in Step 2. This section of the printed circuit is crowded.
Installing R14 first will permit you to mount the adjoining components with ease.
- After mounting the two snap-in sockets on the printed-circuit board as instructed in Step
2, be sure to solder each connection point as instructed. Do not be fooled into thinking that
these connection points make a good electrical connection.
- When connecting R10 to the printed-circuit board in Step 4, do not cut the leads. Otherwise,
wire must be added to reach a connection point later on.
- In Step 4, the instructions call for tinning the negative and positive terminals of the
dry cell. When doing this, be careful not to apply too much heat with the soldering iron. Excessive
heat will damage the cell.
The instructions state exactly how to calibrate the VoltOhmyst. After
calibration, our model was checked against laboratory standards. All scales except the a.c. ranges
were found to be accurate to within 3%. The a.c. ranges were off because the home 117-volt a.c.
power line was used as an a.c. calibration voltage. Unfortunately, home a.c. power line fluctuates
throughout the day. Night voltage could be as high as 122 volts, but when the power demand is
high, the line voltage often drops to 105 volts.
In general, the calibration procedure
permits satisfactory calibration for most servicing purposes. Comment.
The VoltOhmyst assembles without difficulty, works nicely, is fairly rugged, and has ample accuracy
for most practical work. The meter face is large, calibrations are fairly easy to read, and the
knife-edge pointer permits readings with good accuracy. This reliable measuring device will prove
extremely useful in television and hi-fi repair as well as in many industrial applications.
The VoltOhmyst uses a time-proven RCA circuit for optimum results.