Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
Facebook page, please do.
Below are all of the forum threads, including all
the responses to the original posts.
Measures  Post subject: Input Impedance Posted: Mon Mar 01, 2010 6:57 am 
 Joined: Tue Dec 15, 2009 10:22 am Posts: 9  How can I obtain the input impedance of my circuit from the measured Sparameters?
Is the expression Zin=Zo*(1+S11)/(1S11) a general expression, or it is only valid in the cases in which GammaL is zero? I don't obtain the input impedance using this expression. Is GammaL zero occurs only when the load impedance is equal to the impedance of the network? Or does it occur whenever the load impedance is equal to the reference impedance?
Does anybody know which is the expression than ADS uses to obtain the Zin from the S11 parameter? I know this expression works perfectly, but I would like to obtain Zin by myself from the measured Sparameters, not using ADS (I cannot use touchstone files).
Thank you very much

   
david1hitch  Post subject: Re: Input Impedance Posted: Mon Mar 15, 2010 10:35 pm 
 Joined: Mon Jan 04, 2010 1:12 am Posts: 5  The input impedance of any network is determined by the combination of the real and imaginary elements combined , real being the resistive element and the imaginary being the reactive elements . Your results should be in the for z=R+jwl or z=RjwC . I have spent over 30 years dealing with practical examples

   
SunshineDesign  Post subject: Re: Input Impedance Posted: Wed Apr 07, 2010 3:28 pm 
 Joined: Sat Nov 08, 2008 11:35 pm Posts: 30 Location: Ramona, CA  David didn't answer your question, of course impedance is a complex value. But the question was is the formula he posted the general expression for Input impedance from S11 and the answer is yes. Do remember to convert the S11 from dB to linear terms, where it is a voltage ratio. This can be used to convert the S11 measurement of the Zo =50 or 75 ohms of the measurement instrument plane to the Zload of the device. GammaL zero occurs when the load impedance is equal to the reference impedance i.e. when both impedances are exactly matched. _________________ Sunshine Design Engineering Services


Posted 11/12/2012
