Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
Facebook page, please do.
Below are all of the forum threads, including all
the responses to the original posts.
Post subject: Coax Cable Test Help
Posted: Fri Sep 29, 2006 11:11 am
Joined: Fri Sep 29,
2006 11:05 am Posts: 4
Hey everyone !
I have a cable assembly
that I would like to get some test data on.
Instead of the usual Type N to Type N, I have a
daisy chain of Type N to Type N "Tee" to Type N.
I can connect to 3 locations but my VNA only has
2 ports. What could I do to test this assembly
for an accurate VSWR or IL measurement ?
Posted: Fri Sep 29, 2006 4:03 pm
Joined: Sun Aug 03,
2003 2:02 pm Posts: 476 Location: Erie,
You should be able
to terminate one arm of the assembly in a
calibrated 50 ohm load (since that is what a 3rd
VNA port would present) and test two ports at a
I assume you realize that your
described configuration will present 25 ohms
(assuming you are using 50 ohm cable) to all
ports in the assmebly since each port will "see"
two, 50 ohm impedances in parallel.