Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
page, please do.
Below are all of the forum threads, including all
the responses to the original posts.
Post subject: 1-port calibration
Posted: Mon Apr 03, 2006 4:13 am
Joined: Mon Apr 03, 2006 4:06 am
I am doing twice-1-port calibration for a 2-port device mounted on a fixture, due to the fact that I have noe real "Thrugh" even not on the commercial calibration structure. Is there a way to post process this data and get a proper 2-port calibration data that I can then embed into the Agilent ENA?
Thanks for any hints.
R&S RF Application Engineer