•−•  ••−•    −•−•  •−  ••−•  •
RF Cafe Morse Code >Hear It<

# Regarding Cable Loss - RF Cafe Forums

Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views. It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if you would like to post something on RF Cafe's Facebook page, please do.

Below are all of the forum threads, including all the responses to the original posts.

Dreamer
Post subject: Regarding Cable Loss
Unread postPosted: Thu Jun 16, 2005 11:38 pm

Hi all,

I had a question regarding measuring cable loss, How to connect the equipments to calculate the cable loss from one end of the cable to another if you have a Signal Generator, Spectrum Analyser and Power meter. How should I connect the cable to and what are the parameters Should I change to know the cable loss.

Dreamer

Top

Guest
Post subject: Cable Loss Measurement
Unread postPosted: Fri Jun 17, 2005 1:26 pm

Assuming you've got cable and equipment with the same impedances (usually 50 Ohms):

1. Measure output of signal generator with power meter
2. Insert significant length of cable between signal generator and power meter.
3. Measure output from cable
4. a. Cable loss (dB) = dBm(gen) - dBm(cable output)
b. Cable loss (fraction) = Pout(cable)/Pout(gen) powers in W, mw, etc.

Note that some power meters require external termination - you will, of course, have to provide that.

An important point is that the terminating impedance must be equal to the characteristic impedance of the line for this simple method to work.

Remember that cable loss is a function of frequency, so measure at the frequency or frequencies that you're interested in.

Good luck.

Top

Dreamer
Post subject: Regarding Cable Loss
Unread postPosted: Fri Jun 17, 2005 1:45 pm

Thank you very much,

Both impedances are at 50 ohms also you have told to measure signal generator with power meter, so mean just input some power from gen into meter and check the reading on the power meter? for that frequency.

Thanks,

Dreamer.

Posted  11/12/2012
Custom Search
More than 10,000 searchable pages indexed.

#### Your RF CafeProgenitor & Webmaster

... single-handedly redefining what an engineering website should be.

Carpe Diem!
(Seize the Day!)

5th MOB: My USAF radar shop

Airplanes and Rockets: My personal hobby website

Equine Kingdom: My daughter Sally's horse riding website