Noise from a Class C Amplifier - RF Cafe Forums
Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views. It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if you would like to post something on RF Cafe's Facebook page, please do.
Below are all of the forum threads, including all the responses to the original posts.
Post subject: Noise from a Class C Amplifier Posted: Fri Jan 27, 2006 10:37 am
This might be a simple question, but I need some assistance. I am looking at the noise generated by a Class C amplifier (to keep things simple, lets assume it is has a 10 watt amplifier) and how to measure it.
Also I would like to know if anyone has some suggestions about how to analyze the noise (instead of measuring it, can I perform some calculations and get an approximate answer?)
Finally, has anyone tried to simulate the noise from a Class C amplifier?
Any information will be appreciated - websites, magazine articles, books, etc.
Post subject: Posted: Sun Jan 29, 2006 11:54 am
Joined: Mon Jun 27, 2005 2:02 pm
You can measure the output noise of the PA by using Spectrum Analyzer. You first have to measure the output noise when the PA is off, this result will be used as reference point, you have to know the temperature at which it was measured. Then turn on the PA and notice that the noise floor is increased. Make these measurements at the same RBW.
The difference between the noise floors at both cases is the gain of the noise + PA´s Noise Figure. Of course that the noise is also dependent on the BW of operation (RBW of the SA in this case) and the temperature according to: Pn=kTB
After this calculation, you can deduce the self-contribution of the PA to the noise measured.