Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
page, please do.
Below are all of the forum threads, including all
the responses to the original posts.
Post subject: How to measure Phase Noise and Power Spectral Density
Unread postPosted: Thu Dec
30, 2004 5:04 am
How to measure Phase Noise (dBc/Hz) and Power Spectral Density (dBm/Hz) using Spectrum
Unread postPosted: Thu Dec 30, 2004 2:07 pm
In most modern Spectrum Analyzers there is a built in feature that measures the Phase
Noise and plots it in logarithmic graph: Phase Noise vs. Frequency.
However, if you are using older
Spectrum Analyzer, you have to bring the carrier to the center by using MARKER>CF, then placing the reference
marker on the carrier and moving the delta marker to the desired offset. The reading would be the Phase Noise (dBc/Hz).
You have to remember to add the attenuation of the RBW filter= 10 log RBW to this reading. This filter further
attenuates the Phase Noise. I am not sure if you can get the Spectral Density VALUE with Spectrum Analyzer. But
you can measure the outpot power by Power Meter and make the calculation based on the Phase Noise result.
Post subject: PSD measurement
Unread postPosted: Fri Dec 31, 2004 4:54 am
PSD is used when measuring continuous broadband noise, and normalizes the power to an equivalent bandwidth of
1 Hz, irrespective of the actual bandwidth of the filter being used. For example, if a signal is measured at -93
dB in a 10 Hz bandwidth, then the spectral density would be -103 dB (in a 1 Hz bandwidth). But make sure that the
sweep time in a given RBW = Span / RBW in sec. This makes it possible to compare noise measurements made with
different bandwidth settings of the spectrum analyzer.