Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
page, please do.
To the left are all of the forum threads, including all the responses to the original
Post subject: HP87512/VNA
Unread postPosted: Thu Nov
11, 2004 4:32 pm
I have a question for those who have 'worked'
closely with the equations involved in the VNA, and particularly for
the calibration procedure.
At work we have a HP8751A VNA (5Hz-50MHz)
working with a
reflexion/Transmission Test Set HP87512. As I'm always
loking for 'the way it work', I opened it, thinking it was a directional
bridge made with 50ohms resistors (Wheastone bridge) with the design
adapted to work from 5Hz to 500MHz.
But it was a great surprise for
me to see that the 'bridge' from HP,made of resistors, have nothing
to do with a Wheastone bridge (from what I could understand !!). With
the schematic from HP, I can't manage to find a 'directivity' behavior
of the 'bridge' made of resistor. As the HP8751A work with a S parameter
HP87511A or B, I know that the 'bridge' is not a Voltage/Current method
of measuring the reflexion parameter. And if I look on the 'mathematics'
for the calibration of a VNA (TRL or SOL methods,...), there is always
a correction factor for the unperfect directivity of the directional
So is it possible to obtain the correcting factor even
with a nondirective bridge, or a SOL method is able to accept a nondirectional
bridge ???? It seems that the signal from the Output connector is 'split'
with diferent levels to the Reference Port and the A port...
in a real confusion ...!!!
If you have any idea about the way
the HP87512 work with the HP8751A, I would be pleased to read you on
the subject (or if you know a link with the description of the way it