Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
page, please do.
Below are all of the forum threads, including all
the responses to the original posts.
Post subject: dBm to Vrms
Posted: Sat Oct 28, 2006 12:38 pm
Joined: Sat Mar 19, 2005 3:20 pm
I generated a 50 MHz, -10 dBm sine-wave using a Rohde & Schwarz RF signal generator. I checked this using an Agilent spectrum analyser and, as you'd expect, I get my tone at 50 MHz with peak amplitude of -10 dBm. However, when I input my RF signal to a TDS 210 Tecktronix Digital Oscilloscope, the peak-to-peak voltage is 336 mV and the Cyc RMS voltage is 117 mV. Now, on the TDS210, Cyc RMS calculates the RMS voltage over one complete cycle of the waveform.
When I tried to calculate this I got a different answer:
P = -10 dBm, which is equivalent to 100 uW
Now, P = V^2 / R
I've taken R = 50 ohms
Therefore Vrms = sqrt(P x R)
That is: Vrms = sqrt(100 uW x 50 ohms)
So, Vrms = 70.7 mV
This is different to the 117 mV measured on the oscilloscope??
Posted: Mon Oct 30, 2006 4:48 am
Joined: Fri Feb 17, 2006 12:07 pm
Location: London UK
3 possibilities come to mind
1) error in calibration of the scope, particularly at the measurement frequency. If the Y video amp was 3 dB down at the measuring frequency, there is a 3dB error immediately
2) the impedance was higher than 50 ohms.
3) presence of harmonics in the generated waveform.