RP-TNC/f calibration 50 Ohm load ? - RF Cafe Forums
Because of the high maintenance needed to monitor and filter spammers from the RF Cafe Forums, I decided that it would
be best to just archive the pages to make all the good information posted in the past available for review. It is unfortunate
that the scumbags of the world ruin an otherwise useful venue for people wanting to exchanged useful ideas and views.
It seems that the more formal social media like Facebook pretty much dominate this kind of venue anymore anyway, so if
you would like to post something on RF Cafe's
page, please do.
Below are all of the forum threads, including all
the responses to the original posts.
Post subject: RP-TNC/f calibration 50 Ohm load ?
Posted: Wed Jul 27, 2005 4:12 pm
Joined: Wed Jul 27, 2005
Hi, I am trying to eliminate as many interconnections
as possible to run SWR test of 2.4 GHz omnidirectional
'rubber-duck' WiFi antenna. I am using Agilent E5071B
NA. Its port has N/f interface. Antenna has RP-TNC/f
interface. My ideal situation would be to use just
one N/m to RP-TNC/m adapter (I got one). I need
to calibrate Network Analyzer with 50-Ohm Load,
Short, Open with that adapter attached.
anybody know where can I get those calibration tools
with RP-TNC/f interface? Do they exist? Is it possible
to make them on my own?
Any ideas will be
More than 10,000 searchable pages indexed.
Your RF Cafe
single-handedly redefining what an engineering website should be.
Progenitor & Webmaster
(Seize the Day!)
My USAF radar shop